Each Steriking® transportation carton is labelled with the information about the “expiration” date and the manufacturing date.
No collapsing of performance of the product will take place after the recommended expiry date if the storage conditions have been according to the recommendations. It is recommended that the Steriking® products are kept in the original, closed transport cartons and stored in dry and clean conditions protected from direct sunlight and excessive moisture and heat.
The shelf life of a packaged and sterilized item is regarded as event-related rather than time-related. The shelf life strongly depends on the storing and handling conditions of a sterile product. An event must occur to compromise package content sterility. Determination of sterility maintenance time of a final sterile product is in responsibility of the final pack manufacturer that has assembled the sterile barrier system or packaging system in the healthcare facility.
The shelf life of the sterile barrier system and packaging material is an important criterion for this evaluation, but not the single one. Other factors like storage and transportation means and conditions, the amount of sub-handling processes, sterilization parameters and process fulfillment, the properties of the packed item etc. have to be taken into consideration for the final decision about the sterility maintenance time of the final product.
All sterile barrier systems need to be inspected for integrity prior to their use. If the package as sterile barrier system is damaged, the content must be repacked and resterilized.
On the basis of above and because of great number of unknown variables which can be involved and cannot be affected by us as a producer of the sterile barrier system, we may notify the following maximum storing times for the sterilized hospital items:
- Steriking® regular heat sealable peel pouches and reels 12 months
- Steriking® self sealable peel pouches 6 months
Earlier clinical tests made with Steriking® peel pouches and reels have indicated that theoretically and in single test conditions integrity of the sterile package can be maintained up to several years, if proper handling and storing conditions are provided by the manufacturer of the final pack.